The Teseq CAS 3025 burst/EFT pulse verification kit has been specially designed to comply with the new requirements of IEC/EN 61000-4-4:2004 to enable EMC engineers to verify their burst/EFT test generators periodically.
Annual calibration and periodic verification
The annual calibration of test equipment recommended by most quality systems (ISO 9000, ISO 17025, etc.) has to be considered as a validation of all measurements done since the last calibration.
Many EMC standards call for a verification of the test equipment before and after every test session. If the verification shows different results, no valid test results can be assumed and the test equipment has to be re-calibrated.
Teseq CAS3025 Features
- Verification/calibration kit for IEC/EN 61000-4-4, amendment 2
- For peak voltages up to 8.8 kV
- INA 265B: 50 Ω termination
- INA 266A: 1 kΩ termination
- Compatible with all burst/EFT generators
Teseq CAS3025 Specs
|INA 265B termination/attenuation (typical)
||50 Ω; 1:1000
|INA 266A termination/attenuation (typical)
||1000 Ω; 1:2000
|Max. input burst peak voltage
||8800 V peak
||-10 to +40°C